DIN Panel mount - Temperature Indicator
with very large digits
The D040 is a straight forward, panel mount temperature indicator. The measuring unit to be displayed is simply selected through an alfa-numerical configuration menu. No adhesive labels have to be put on the outside of the enclosure: a weather proof and user friendly solution! The configuration of the Span, off-set and number of decimals is done through software functions, without any sensitive dip-switches or trimmers.
- Unique, robust IP66, IP67 (NEMA Type4X) panel mount front enclosure made of die cast aluminum, allowing even big jets of water and total immersion.
- Familiar easy operation with the Fluidwell "Know one, know them all" configuration structure, saving time, cost and aggravation.
You will receive a email with the download link soon.
- Displays actual temperature and measuring unit.
- Very large 26mm (1") digits.
- Piegraph indication: ten segments.
- Number of digits for temperature: 5½.
- Selectable on-screen engineering units: °C - °F - K.
- Auto backup of all settings.
- Ambient temperature: -40°C to +80°C (-40°F to 176°F).
- LED backlight option.
- Easy configuration with clear alphanumerical display.
- Aluminum DIN front panel of 144 x 72mm (5.67” x 2.83”) according DIN 43700 / IEC 61554.
- Various accessories are available for this product.
- The D040 fits in applications where a basic temperature measurement display is required without temperature monitoring.
- The D-Series is a DIN-sized display and the better alternative for your existing, not waterproof, front panel mount indicators in extreme weather outdoor applications or e.g. in food industries where working environments are often cleaned with powerful water jets.
- (0)4 – 20mA.
- 0 – 10V DC.
- Basic: 8 - 30V DC (no real sensor supply).
- Long life Lithium battery.
- 4 – 20mA input Loop-powered.
- 24V AC/DC + sensor supply.
- 115 – 230V AC + sensor supply.
- 8.2, 12 & 24V DC.
- For Intrinsically Safe applications we recommend the F-Series.
Questions and answers
No questions have been asked about this article yet.Ask a question
Your question has been forwarded to our support team.
We will answer it as soon as possible.